Figure 45. Pettersson small chain hydrometer, designed by Professors Otto and
Hans Pettersson in 1917. This instrument eliminated difficulties associated
with the surface tension of the sample as well as variations of volume. A small
metal chain helped achieve a balance by compensating for the buoyancy of the
float as a function of the density of the liquid.